This article provides you a example radioactive level instrument inspection and test plan. This is a draft and must be edited based on your purchase order specifications.
|Radioactive Level Instrument Inspection and Test Plan Task||Vendor||TPI Inspection||Client|
|Radioactive level instrument mill test report||H||R||R|
|Radioactive level instrument Calibration test||H||W||R|
|Radioactive level instrument Performance test including hystersis||H||W||R|
|Radioactive level instrument Insulation resistance test||H||W||R|
|Radioactive level instrument Final visual / Dimension Inspection||H||H||R|
|Radioactive level instrument Documentation review prior to release(Final activity of Technical Inspection)||H||H||R|
|Radioactive level instrument Pre-shipment Inspection||H||H||R|
H: Hold Point implies that relevant production activities shall not proceed until the continuation of work is permitted by Purchaser. (Notification req'd)
W: Witness point implies that Purchaser intends to witness the designated inspection feature. If Purchaser decides not to witness the relevant feature, production can proceed without permission of Purchaser. (Notification req'd)
S: Witness, but spot check basis. Initial operation will be witness point and subsequent operation will be witnessed at discretion of Purchaser considering the results of previous inspection. (Notification not req'd : Randam Inspection)
R: Review inspection & test records (Notification not req'd)
General Note Radioactive Level Instrument ITP:
1) Radioactive Level Instrument witness/hold point by Code Inspector, when applied by Code /Local Regulation, is to comply with its requirements and to specify in Vendor's ITP.
2) Type of Inspection by Radioactive Level Instrument Purchaser may be changed according to equipment criticality and vendor QC evaluation at job stage.
3) When requested in the requisition, detailed Radioactive Level Instrument ITP shall be submitted by Vendor.
4) Monitoring/Observation of Radioactive Level Instrument Vendor shop daily routine works (i.e. Storage of materials, Adherence to approved procedures, Testing tool calibration check, workmanship, cleanliness and etc.) shall be done by Purchaser during inspection visit.
5) 100% Radioactive Level Instrument inspection which is covered by the combination of witness / spot check / record review inspection & tests listed above and vendor's original QC activity shall be confirmed by Purchaser's inspector prior to shipment.
6) Unless otherwise specified in Radioactive Level Instrument REQ., the sampling Q'ty of " S (Spot Witness Check) " shall be as follows:
3 to 20→3Pcs(all if total 2 and less) , 21 to 50→5Pcs, 51 to 100→10Pcs,101 to 200→20Pcs, 201 to 300→30Pcs, 301 to 500→50Pcs
Minimum 25 pcs or more of the commodity shall be arranged for H/W/S inspection and test at one time inspector's visit as much as practical.
7) Radioactive Level Instrument Final Visual Inspection/Documentation Review/Pre-Shipment Inspection may be combined within one time visit.
Did you find this article useful? Click on below Like and G+1 buttons!